Tables 2—2 below gives examples of the most frequently used configurations and kV values. It allows selection of the optimum kV, and filter combination in the selected priority. According to the composition of the breast to be examined, select a configuration with suitable kV values. Align the two different—sized pins of the new bulb with its socked and push the bulb upwards. Plug for connecting cable to generator For a detailed description of the control console use, refer to Section 3.
If you become aware of liquid entry, disconnect the electrical supply and have the equipment checked by qualified service personnel before returning it to use. To unlock the magnification plate, press the lever located on the left of the mag. Readout for messages and keys selections 5. The manufacturer is not responsible for any interference caused either by the use of interconnect cables other than those recommended or by unauthorized changes or modifications to this equipment. Manual adjustment is via thumbwheels located one on either side of the compression paddle arm.Next
Equipment parts should be sprayed with a fine mist applicator or wiped with a wet cloth or sponge as directed by the instructions for use provided with the germicide. A readout, located at the bottom of the column indicates: Examination arm angulation in degrees — ° , Compressed breast thickness in millimeters — mm , Compression force in deca Newtons — daN. Full field Included in the basic configuration 14 x 14 Green Small 0. Where simultaneous biplane examinations are undertaken with the x-ray units, the supply conductors and overcurrent protective devices shall be 100 percent of the momentary demand rating of each x-ray unit. A buzzer sounds at the end of exposure.
Data Flash for recording and printing all exam and patient data directly onto the film. It takes full advantage of the digital technology from on-screen image display non-diagnostic softcopy reading to networking, filming and archiving. The operator should use the cassette holder. If skin preparations contain high alcohol concentrations, allow sufficient drying time before applying compression; Do not expose equipment parts to steam or high temperature sterilization; Never allow liquids to enter the internal parts of the equipment. Check resistance of belts in decompression. The compression paddle automatically moves up to the pre—programmed autodecompression height. Insertion: Slide Bucky or cassette holder on film holder assembly.Next
Compression release button: Press to release compression. Exposure Disable lamp When red lamp is illuminated, exposure is disabled. Check field covered by centering light. At end of exposure, the kV readout displays the kV values selected. If required, the system can provide as well, high image quality radiography of specimens and of the hand and foot. It also allows easy access to the light bulb for easy replacement see Chapter 3 — Section 1.Next
Five positions are available to suit patient anatomy. The machine will will do the exposure. The kV readout displays the first letter of the code of the selected mode C, S, or D and the mAs readout displays the mAs values output. Yes General: Is the device under confiscation? This Increase the kV value and reduce the message only appears in mAs value. Refurbished Senographe 2000D allows you to protect your budget and still get advanced mammography technology.
Vor Inbetriebnahme dieses Apparats sollte sich das qualifizierte und befugte Bedienungspersonal mit den geltenden Kriterien für den gefahrlosen Strahleneinsatz durch sorgfältiges Studium des Hefts Nr. Compression release Button Note: The symbol located on the control panel means that you must read the appropriate chapter in the operator manual before taking any action. The mAs values are given by: mAs T1 idt T2 Where T1 is the time at which the high voltage reaches 75% of its maximum value, and T is the time at which the high voltage returns to 75% of the maximum value. It also means that you can ignore the message if a different, less dense, breast is to be imaged next or if different technical factors are selected. However, it is possible, for special purposes such as physicist testing, to disable this interlock.Next
Consult the point of purchase or the service representative for further suggestions. Power the equipment from a source different from that of the affected device. Exposure Enable indicator lamp 16. However, it is possible for special purposes such as physicist testing for example to disable this interlock see chapter 3 for a detailed explanation. Density correction is used only when an optical density change in the image is required. An exposure during which the X—Ray exposure is made. Yes General: Are there any visual or non visual damages? Guide Rail and Compression system including compression paddle and compression paddle support.Next
This would prevent the exposure interlock feature from functioning correctly. Remove the packing of the new bulb. The light centering switch is also located at the top and bottom of each handrail. Unknown X-ray Tube: What is the manufacturing date of the X-ray tube? The seven remaining characters are also identification characters. A reduction in the kV value improves film contrast but increases the radiation dose to the patient.Next